The Pocket Colorimeter II™ has two manuals. The Basic and Full User Manuals can be found below. All Pocket Colorimeter™ II instruments. This symbol, if noted on the instrument, references the instruction manual Hach Pocket Colorimeter™ instruments* are low-cost, high-quality filter photometers. Consult HACH’s Pocket Colorimeter II brochure on ArchiExpo. Page: 1/4.
|Published (Last):||5 November 2011|
|PDF File Size:||15.7 Mb|
|ePub File Size:||19.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
Refer to the appropriate message information below colorieter determine what the problem is and how it can be corrected. Repeat this sequence until the S0 concentration value is entered.
Editing a User-entered or Factory Calibration Curve 1. Page 62 Standard Calibration Adjust, continued 7. Wavelength-specic models without factory calibration are also available for programming custom methods.
At least two data pairs concentration and absorbance are required. The Standard Calibration Adjust can be used to meet regulatory requirements.
Instrument Operation, continued Battery Installation Figure 1 on page 2—7 provides an exploded view of battery installation. Remove power from the Pocket Colorimeter instrument by removing one of its batteries to verify that it is or is not the source of the interference. Page 71 User-Entered Calibration, continued 5.
Many are based on EPA-approved methods. Place the blank in the cell The colorineter should indicate holder with the diamond Samples must be Used to enter a predetermined calibration hxch.
May not develop full color or color may fade instantly. Cover the blank with the instrument cap. Adjust sample pH to 6—7. If error recurs, re-zero the instrument.
Each instrument comes in a ready-to-use kit that includes a manual, sample cells, and a sturdy custom carrying case. Simply zero the instrument with a blank, colroimeter the reacted sample, and read the result.
Warranty In the event that a defect is discovered during the warranty period, Hach Company agrees that, at its option, it will repair or replace the defective product or refund the purchase price, excluding original shipping and handling charges. Forums Documents Parts Videos News. Page 49 Section 2 Instrument Manual 2— Press to return to the measurement mode.
Hach POCKET COLORIMETER II Instruction Manual
User-Entered Calibration, continued 5. Printed in the U. Enter text from picture: Enter the concentration value and absorbance value of the first data pair S0, A0. On the basis of strict liability or under any other legal theory, in no event shall Hach Company be liable for any incidental or consequential damages of any kind for breach of warranty coolorimeter negligence.
Pocket ColoriMeter with standards PCII
To ensure the protection provided by this equipment is not impaired, do not use or install this equipment in any manner other than that which is specified in this manual. Remove the sample blank.
A chlorine standard may be prepared and used to validate the calibration curve using the Standard Calibration Adjust feature see page 2—13 for more A user calibration or a user-prepared chlorine mamual may be required by a regulatory official or agency. The curve is generated by a point-to-point straight line segment between each standard used. Place the blank into the cell holder, with the The arrow should indicate the Samples must be Note Many are based on EPA-approved methods.
When all the pockket data has been entered, press twice to return to MENU the measurement mode. Place the blank in the Cap.
Hach Pocket Colorimeter™ II for Chlorine or Manganese | Pine Environmental
Insert the reagent blank or yach water into the cell holder. Use of the factory calibration is recommended unless the user is required to generate a calibration. The curve can extend from 0 to 2.
Chlorine, Free And Total, Lr For best results, dedicate a set of cells to each free chlorine and total chlorine test.